Payment & Shipping Terms:
|Material:||Stainless Steel||Color:||Steel Color|
|Size:||900x 800 X 1700 Mm Or Customized||Usage:||High Temperature Air Accelerate Tes|
|Power Supply:||220V 50Hz||Temperature Range:||135 Degree|
Equipment use and characteristics:
1. Adopt the dual structure of imported high-temperature solenoid valve to minimize the use failure rate.
2. Independent steam generation room, to avoid the direct impact of steam products, so as to avoid local damage.
3. Lock the power structure of the door, and solve the defect of the locking difficulty of the first-generation product disc handle.
4. Cold air before test;In the experiment, the design of emptying (the air in the test barrel) increases the pressure stability and reproducibility.
5. Long time experiment operation time, the long time experiment machine is operated for 400 hours.
6. Water level protection, through the test of water level Sensor in the laboratory.
7. Design of tank pressure, casing pressure resistant (140 ℃), 2.65 kg, in line with the water pressure test 6 kg.
8. Two-stage pressure safety protection device, adopt two-stage combination controller and mechanical pressure protection device.
9. Safety protection row button, emergency safety device, two-stage automatic pressure release button.
1. The performance
1.1 setting temperature: normal temperature ~ +135 (saturated vapor temperature)
1.2 humidity range: saturated 100 % vapor humidity.
1.3 humidity control stability: plus or minus 1%RH.
1.3 use pressure: 1.2~2kg.
1.4 time range: 000 Hr ~ 999 Hr.
1.5 pressure time: 0.00 Kg ~ 1.04 Kg/cm2 about 45 points.
Temperature fluctuations uniformity: 1.6 + / - 0.5 ℃
1.7 temperature display accuracy: 0.1
1.8 pressure fluctuation uniformity: + 0.02Kg.
1.9 humidity distribution: + 5%RH.
2. Material of test box:
2.1 test box size: PCT-150 with 350 mm x L500 mm
2.2 total external dimensions: about 900x 800 x 1700 mm (W * D * H) vertical.
2.3 inner barrel material: stainless steel sheet material (SUS# 304 5mm)
2.4 external barrel material: stainless steel sheet metal spray paint.
2.5 insulation material: rock wool and hard polyurethane foam insulation.
2.6 steam room heating tube: finned heat pipe seamless steel tube electric heater (surface plating platinum, anticorrosion)
2.7 control system:
A. Adopt daily RKC microcomputer to control the saturated vapor temperature (PT-100 platinum temperature sensor).
B. time controller adopts LED display.
C. Display pressure gauge with pointer.
D. microcomputer P.I.D automatically controls saturated vapor temperature.
E. Manual inlet valve (automatic water filling function, continuous test without stopping).
2.8 mechanical structure:
A. Circular inner box, stainless steel round test case structure, conforms to industrial safety container standard, can prevent the dew drop design in the test.
B. Circular interior lining, stainless steel circular interior design, can avoid the direct impact test of steam latent heat.
C. Precise design, good air tightness, low water consumption, and can operate at 400Hrs continuously at a time.
D. Patented packing design enables the door to be more closely integrated with the box body, which is completely different from the traditional extrusion type, which can extend the packing life.
E. automatic security protection, abnormal cause and fault indicator light display.
2.9 security protection:
A. imported high temperature sealed solenoid valve with double loop structure to ensure no leakage of pressure.
B. The whole machine is equipped with overpressure protection, overtemperature protection, one-button pressure relief, manual relief of multiple safety devices, to the maximum extent to ensure users' use and safety.
C. Reverse lock device. When there is pressure inside the test chamber, the door of the test chamber cannot be opened.
3. Other accessories.
3.1 test set 1 set.
3.2 sample plate
4. Power supply system:
4.1 the fluctuation of the system power supply shall not be greater than or minus 10.
4.2 power supply: single-phase 220V 20A 50/60hz.
5. Environment & facilities:
5.1 the working environment temperature can be used for 5 ~30.
5.2 experimental water: pure water or distilled water.
Place of OriginGuangdong, China (Mainland)
Usage:Universal Testing Machine
Temperature range:135 Degree
Refrigerator:Origin imported full sealed compressor from European and USA market
Chamber:350 mm x L500 mm
Inside:SUS#304 stainless steel material
Power supply:220V 20A 50/60Hz